RD8100 RF Marker locators are capable of recording up to 1000 survey measurement records to internal memory. When a measurement is taken and saved, the RD8100 Marker locator will store the most relevant measurements taken and information, such as:
These measurements, taken at specific points indicated by you, can be used to report and map all your findings.
If the locator is a GPS model or if paired to an Android device running RD Map™ or if paired to a Pocket PC (or PDA) with GPS, running a compatible application such as SurveyCERT™+ the locator will append position and time information to the data.
NOTE: The internal GPS module needs to be switched on and connected to the GPS satellite system.
WARNING! Do not store more than 1000 measurements, as this will reset the counter and start overwriting existing measurements. It is recommended to regularly retrieve the survey measurements data and erase them from the locator,
To save survey measurements, press the key.
To achieve accurate results the locator must be kept as still as possible during the saving process.
The locator will always save measurements to internal memory. If Bluetooth is switched on and the Bluetooth survey measurements transmission are active the locator will attempt to send the saved measurement to a paired device running a compatible application.
If your paired device is out of reach, or if its Bluetooth features are disabled, the locator will display an error code. To avoid these errors ensure that your paired device is within range and correctly paired.
NOTE: A flashing depth and/or current reading display means that the measurement is poor and should be taken again. Poor readings may be caused by nearby conductors or sources of electromagnetic interference.
To configure Bluetooth survey measurements transmissions:
The RD8100 Marker locator allows you to delete all measurements. Erasing the survey measurements will wipe the RD8100 memory and is usually recommended when you begin a new survey and you have saved your previous measurements.
WARNING!: Erasing measurements cannot be undone! Proceed with caution!
To erase all stored measurements:
Stored data can be transferred wirelessly to a compatible PDA running Radiodetection’s SurveyCERT+ app or a PDA compatible application.
The Radiodetection SurveyCERT+ PDA app can read survey data for post survey analysis.
NOTE: SurveyCERT+ is compatible with Bluetooth enabled PDAs running Windows Mobile 6.x operating systems.
Stored data can also be transferred using the USB connection to the RD Manager software package to be analyzed by the software’s built-in Survey measurements capabilities.
Refer to the RD Manager operation manual for more info on how to retrieve store survey data.
RD Manager’s Survey measurements functionality can be used for post survey analysis, interface to Google Earth® and easy export to GIS / mapping systems.
NOTE: RD Manager is compatible with PCs running Microsoft® Windows® XP, onward.
To send stored data to your paired PDA:
NOTE: When using SurveyCERT+, the RD8100M locator Bluetooth protocol needs to be set to PPP.
To obtain SurveyCERT+ for PDA:
SurveyCERT+ for PDAs and it’s operation manual are available as a free download from the Radiodetection web Site:
To obtain RD Manager: